August 20, 2013 04:42
Does anyone know what error code 226C4 means upon power up of CR 85-X. Please advise or if you have service manual with error codes for this unit
to rate and post a comment
re: AGFA CR85-X
August 20, 2013 07:10
Power supply test of stepper node belt failed. Check the cables and connectors, fuses, power supply, safety interlocks.
The next explications is referent your error in the CR85-X...
Reason : Input buffer module: Power supply test of stepper node Belt failed.
Meaning : No Power supply on stepper node detected.
- power supply not connected.
- fuse blown.
- power supply defective.
Cure : - check cables and connectors.
- check/replace fuse.
- check/replace power supply.
Sentient is the premier resource for buying, selling, repairing & refurbishing pre-owned cosmetic lasers & energy based devices. Top OEM's including Cutera, Lumenis, Palomar, Alma, Syneron & Cynosure in stock now!