Siemens Siremobile Compact C Arm error code 7409
January 26, 2010 04:20
Siemens Siremobile Compact C Arm System. Have error code 7409.
to rate and post a comment
Re: Siemens Siremobile Compact C Arm error code 7409
January 27, 2010 08:01
Error E7409 TV-Iris wrong position
If the error 7400 to 7408 were detected, this error is displayed after each radiation release.
The user must delete this error before radiation can be released.
For more informations, read out the previous error messages.
Check the doserate adjustment.
Check the actual value potentiometer.
Check the actual value potentiometer wires.
H&S Medical can now rebuild your Steris Amsco 3080 & 3085 for about $5,000 using our new propiatary techniques. If your table has become loose and wobbly call us to discuss your solutions.