Ziehm Exoscop 7000 Error 26 during exposure - service manual
January 11, 2011 12:40
Good morning. I need a service manual for a ziehm exoscop 7000 or some insight as to what the error code E26 represents when trying to make an exposure. I can make a 50kv manual exposure?
Is it tube arc??
Any assistance is greatly appreciated.
to rate and post a comment
re: Ziehm Exoscop 7000 Error 26 during exposure - service manual
July 12, 2012 01:46
Did you find out what the error 26 was? Im having the same problem?
re: re: Ziehm Exoscop 7000 Error 26 during exposure - service manual
November 26, 2012 12:34
No, never found out, we left it unplugged for sometime, plugged it back in and now reading error 32, which I was told means batteries on cpu board need replacing, we decided to sale as parts unit
February 27, 2013 04:43
re: re: re: Ziehm exposcop 7000 compact error code 26
July 25, 2012 08:00
It means there was a generator error, could be the generator needs to be recalibrated, or a bad connection to the tube or bad wire withing the cable harness or worse case a faulty, arcing tube. Errors 26 & 29 in most cases means the tube is arcing and needs to be replaced but you could get lucky with a recalibration or checking the connections to the tube. The high voltage is developed at the tube head which is usually at fault...good luck
15,000 service professionals from over 50 different countries have trained in over 35 courses in radiology, mammography, CT, ultrasound, networking, PACS, & DICOM. Classroom & lab facilities to accommodate up to 60 students.