Here is a ProteinSimple iCE3 Protein Analyzer
Made in 2014
Last PM performed in June 2018
System was originally in use by Pfizer and comes with original computer.
New one sells for around $85,000
This ProteinSimple iCE3 Protein Analyzer is in good working order.
Please see the exact video and photos below.
Testing we performed:
The ProteinSimple iCE3 Protein Analyzer passed all tests and was found to be fully functional:
Tested the voltage control on the iCE by connecting a high voltage probe to the electrodes via leads to the meter set to DC voltage.
**Note: with the high voltage probe the meter is scaled down so that 1.50 volts are equal to 1,500 volts measured by the probe**
Tested Voltage accuracy to better than 1% of set point: PASSED
Tested acquisition of a dark scan: PASSED
Tested acquisition of an empty One Image scan: PASSED
Tested acquisition of One Image scans with 2 custom slides: PASSED
Tested acquisition of One Image scan using stacked ND filters: PASSED
iCE valve mode switch test, observed flow through the valve: PASSED
The testing performed on this ProteinSimple iCE3 Protein Analyzer verifies popper operation of the electronics, detector,
Lamp, high voltage power supply and, mode switch valve. Due to this testing you can purchase with confidence knowing this
ProteinSimple iCE3 Protein Analyzer is ready to go.
If any information leaves you with questions please contact us so we can get you answers.
Tested temperature control from 4-40C: PASSED
Ran "Needle calibration" procedure: PASSED
Ran "Sample injection" procedure: PASSED
Ran "Rinse" procedure: PASSED
Ran "Park" procedure: PASSED
This ProteinSimple 720NV Autosampler has been cleaned and has had every servo and actuator ran though their full range without issue.
The pump, needle and flow path were tested using DI water ensuring the system has been rinsed out and is clear of obstruction.
System consists of:
ProteinSimple iCE3 Protein Analyzer
Desktop PC running Windows 7 Pro with iCE CFR software ver4.1
Communication and power cables.
Configuration and Setup:
FC Cartridge (Requires Methyl Cellulose), HT Cartridge (Does not require Methyl Cellulose)(No Cartridges included)
Dimensions: iCE3 (WxDxH) 28.25 x 31 x 60.5cm, 720NV (WxDxH) 65 x 55 x 66cm
The ProteinSimple iCE3 Protein Analyzer with 720NV autosampler (figure 4-28) allows injection of samples into the cIEF cartridge using a positive displacement syringe pump system.
The autosampler can accommodate either 11 mm vials or a 96-well microtiter plate, and provides temperature control between 4-40C.
This ProteinSimple iCE3 Protein Analyzer performs free solution isoelectric focusing in a capillary column and detects focused protein zones using a whole column UV absorption detector.
We zoom out for front view of ProteinSimple iCE3 Protein Analyzer, showing software connection and functions.
Then we show exterior of ProteinSimple iCE3 Protein Analyzer, zooming on tags and connections. Coming back to software to show autosampler function.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.