Checkup failure Siemens Somatom Scope 2017
November 16, 2021 01:56
I just encountered a problem when performing checkup procedure to a Somatom Scope Power from 2017. During the procedure is entering in a loop (repeating last scan until fatal error ). Any idea? Thank you in advance!
to rate and post a comment
JIFFY PAUL AKKARAKARAN
re: Checkup failure Siemens Somatom Scope 2017
November 17, 2021 10:11
Please check the event log. You can get details of the error. If it is aborting scan in warm up cycle then it could be either x-ray tube or the x-ray generator issues. Is aborting in filament adaptation cycle or defective channel correction or in the air calibration cycle?
November 22, 2021 11:26
When you have got this error? When system start filament adaptation or deffective channels? If first - then problem can be with tube. If second- check detector or irs
Your comprehensive contrast injector solution for 10 years to include: on-site service, telephone technical support, depot repair with loaner options, parts sales, injector sales, injector install & de-install, Level 2 & beyond training.