September 09, 2021 11:16
Hi, i'm having a trouble with x Ray system
Continental CXD TM50
When i turn on the system
I can't make a xray exposure
The display :
System flag #95
RPU Link up not established
And the led Hold is on
Does any body can help me with this error?
Thank's Sebastian gatica
to rate and post a comment
re: Continental rad
September 10, 2021 10:51
Check the console power supply, the error if I remember correctly from my Continental School days is the console CPU and the Generator CPU cannot communicate with one another. I have had problems with the PICs on the console cpu failing causing a tx/rx problem. All components are off the shelf items and can be purchased from mouser.com.
As for boards, good luck...Transworld, Trans-Continental, Continental and Trex have been out of business for YEARS and there are no parts available.
I can verify the error tomorrow provided I still have my school books from Continental training.
September 13, 2021 10:39
I just looked through my archives and I have disposed of all Transworld, Continental and Trex Medical documentation.
Not sure if anyone else will be available to assist you. I hate to say it but you are trying to resurrect a dead horse and it should be put out to rest.
Premier Medical Depot closeout auctions take place on DOTmed.com. Bid today on 979 lots of medical equipment. Onsite or virtual inspection available. Please contact auctionservices@PremierMedicalDepot.com or call (405) 365-5401