Error 1579 in OMCP (Integris) Philips Generator
May 30, 2019 09:07
Hello, System is Ok at standby (no error) however at fluo or cine NO images and FNMD says "Action" for a while, then recovery Standby "OK". Error trace shows E1579 ("EXON is missing"). EQ Converter seems inactive (no HF noise), Filament boards are OK. MRC is OK. RDY1 and RDY2 leds are lit (normal) in EZ112 during Xray.
I have replaced EH116, EH102, EZ112 w/o changes.
Any further suggestions?
to rate and post a comment
re: Error 1579 in OMCP (Integris) Philips Generator
July 19, 2019 10:19
See 1576 error; Possibly defective EVK3 relay in powerpart;
Jumper on EZ112 should be in position 2-3 in case of MRC gridswitch!;
ElsMed engineers are located in four different countries. 24 hour response time to any service call. Service agreement provided for end-users range from a 6 month period to a 3 year period and covers all system spare parts.