Error 240/032 Siemens Iconos R100
September 24, 2018 08:18
Someone have an experience on Siemens Iconos R100, I have a code error 240/032.
The table not move no more, but all voltajes on board basic unit is working,
I try to connect with service but message on pc that "not all parts working"
to rate and post a comment
re: Error 240/032 Siemens Iconos R100
September 24, 2018 08:47
rror 240/032 (Iconos R100)
Basic unit: Safety breaker sticks when system is switched on.
Unit is not operational. Safety breaker sticks.
- Switch the system off and back on.
- Check the safety breaker; if necessary, replace it.
re: re: Error 240/032 Siemens Iconos R100
September 25, 2018 09:34
all safety breakere stick is good, all voltajes is present but the error is on table,
only I found the voltaje on U20, U7 is 210VAC not 240.
does this is the normal?
September 26, 2018 09:51
Does the safety breaker sticks is K1 on basic unit?
or K1 on A1 D2 on SPOR FILM DEVICE
Basic unit: safety breaker deenegized.
- Emergency stop switch pressed.
- Safety breaker has responded (unit has moved into the safety circuit).
- Limit switch defective or a cable break
- 24 V~ for the safety breaker missing
- 24 V~ for the safety breaker missing: test points A4.D31.XP14 versus XP2; F10 fuse on the A4.T1 transformer OK? (thermoswitches T1.S1 to T1.S3 OK?)
- Move the unit out of the safety circuit.
- Check limit switches and cabling.
September 26, 2018 09:37
i check all system but on D1 B10 red light on and 240/032 show me.
some times the system turn on good no error, but many times the control appear error 240/032
Tekyard Medical closeout auctions take place on DOTmed.com. Bid today on 310 lots of medical equipment. Inspection is by appointment only. Please email at email@example.com for an appointment.