GE LIGHTSPEED PRO 80KV AND 100KV RING ARTIFACT
November 18, 2019 10:39
i have lightspeed G.E 4 slice pro,
during Scanning using 80KV and 100KV Fa
factor "a ring artifact appeared on the image".
Using 120KV image is in good condition, no artifact
any idea about this matter, if anyone can help me what procedure i need to do?
to rate and post a comment
re: GE LIGHTSPEED PRO 80KV AND 100KV RING ARTIFACT
November 19, 2019 10:26
Hi! Do you check detector? Das tools?
November 19, 2019 12:03
You need make calibrations
November 22, 2019 01:00
Restore System State and perform a FastCal, if this didn't work check the detector running DASTools
All parts available upon request. Please include part number and description for your request. Warranty available. Call 813-925-1314 or email at Sales@Elsmed.com