GE Signa Profile 4B _TCU problem
July 19, 2019 10:40
I reinstalled SW on Signa Profile and Now I have error
message : Temperature Controller Communication error detected. Communication time out--Error:2240982/
Before SW reinstallation there was no problem with TCU. It is working and shows target temperature 32.5 now.
Can someone help me with this case?
to rate and post a comment
re: GE Signa Profile 4B _TCU problem
July 21, 2019 01:00
Info files restored properly?
July 21, 2019 01:08
re: re: GE Signa Profile 4B _TCU problem
July 22, 2019 04:07
I am not sure.There was MO disk with saved info and I used it. No more info in service manual: How check SW or HW.
I noticed one message during Solaris boot--ifconfig: SIOCGFFLAGS: vd0: no such interface! Does this message refer to my problem?
July 22, 2019 04:06
I don't think so. hme0 and vd0 are network interface devices.
May be is Bit 3 get problem, this is fiber Card in SunPC and other fiber Card in TPS, No need entangled in TCU
July 23, 2019 10:33
Your Bit3 cards are fine on Sun PC and TPS chassis.
Dear Steven, how to check, is there problem with Bit3 or not?
I am not good at Solaris systems. Can you recommend something?In other side I am sure that TCU error appeared after SW reinstallation!
re: re: re: GE Signa Profile 4B _TCU problem
July 24, 2019 12:00
I have encountered this problem before,and try many thing like change TCU, I am lucky because has other same type MRI test, first time We exchange the Host and found the Bad Host will be your problem, so we focus to Host, and change new fiber card.
You must to buy a new fiber card. Good Luck
re: re: re: re: GE Signa Profile 4B _TCU problem
July 25, 2019 10:26
Thank you, I will try.
Injector Support & Service (ISS) offers telephone technical support,
depot repair services, loaner injectors, parts identification and sales, PM tools
and PM parts kits for medical contrast injectors.