Lorad M-IV Mammography System Error: 34
January 19, 2017 06:04
Please our Lorad M-IV Mammo displays Error 34. Any idea on what could have caused this and how to resolve it?
I will also appreciate to have the service manual for the Mammo (email@example.com).
to rate and post a comment
re: Lorad M-IV Mammography System Error: 34
January 23, 2017 07:57
34 Displayed Message: System Error 34.
Error: Focal Spot Relay Fault.
Notes: The condition of the focal spot relay on the filament control board does not match the
condition the generator processor has selected.
Check relay and troubleshoot the control signal on the filament board signal comes from the microprocessor board.
Try to reload backups I had this issue once due to corrupt calibration files
re: re: Lorad M-IV Mammography System Error: 34
October 16, 2020 10:29
hello ..do you have the service manual / diagrams for this system? I need a copy asap.
getting error 62 (motion) when doing a LMAO using either bucky
carlos enrique salazar gea
January 23, 2017 07:59
System Error 34 mA Control Focal spot selection fault Check the focal spot relay on the
Filament Control board. Replace
board if necessary.
January 23, 2017 08:03
Hi, Dophs. Send me your email. Thanks
January 13, 2020 02:05
please share me details to solve this error
October 16, 2020 10:31
share me details to solve this error thanks
Family owned since 1919, for your next medical move, rely on the specialists at Brandon Transfer & Storage. For references from the medical community & a free moving estimate, call us at 561-683-5000 and we care! Visit www.brandontransfer.com Today!