Philips HD 11 SERVICE MANUAL
April 18, 2017 04:56
I have an error
An internal error and the machine shall restart
Could you please send me a copy of
Philips HD 11 Service manual on my email
thanks a lot
to rate and post a comment
re: Philips HD 11 SERVICE MANUAL
April 19, 2017 08:18
Make sure the filter under the machine is clean. If it's dirty the CPU will over heat and shut down. Had that problem a few months back and cleaning the filter took care of issue. Also make sure all the fans are working.
April 19, 2017 08:19
Yes I will send you
April 19, 2017 08:21
I sent the information about PS - You should check it!
April 19, 2017 08:23
Please repair proces start with Maintenance- cleanup from dust PC and ATX power supply, e-Box and System Power supply.
Them recconect all boards and cables, speciale betwin APIO (PC)and BPAP (e-VBOX).
Check PC ATX power supplay Voltages, Hard Disk for error.
After all work`s check machine normal Boot and if not start and have problem, wtite please repot there.
HD11 is same as EnVisor M2540A. Service manual You can find with google "Service manual M2549A" or "Service manual HD11".
re: re: Philips HD 11 SERVICE MANUAL
April 20, 2017 10:18
thanks alot for your answer
The power supply ok
all voltage is normally
I do not know what else can happen
it 's crazy error
re: re: re: Philips HD 11 SERVICE MANUAL
April 21, 2017 10:14
HD11 have 2 power supply - System power supply in e-BOX and PC ATX power supply. All voltages OK?
Did You check HDD, reconnect all cable connector`s and board`s?
re: re: re: re: Philips HD 11 SERVICE MANUAL
April 24, 2017 10:15
Thanks for your answer
The power supplies are good both
The hard disk is good
I will look at the cables and tell you
thanks a lot
We offer full service contracts, PM contracts, rapid response, time and material,
camera relocation. Nuclear medicine equipment service provider since 1975. Click or call now for more information 800 96 NUMED