siemens arcadis varic error
April 07, 2021 10:09
hi, i have a troble with a c arm siemens arcadis varic
During power up ADC tests failed
if the failure occurs frequently , inform service
if somebody knows about this error i will aprecciate the support.
to rate and post a comment
re: siemens arcadis varic error
April 08, 2021 10:42
Explanation: a) During power up, ADC tests fails
Action: a1) Check the voltage supply M14 (+/-15V)
a2) If the fault doesn't come if the plug D1 X1 is disconected
please check the connection to the TV iris.
a3) Check cables from voltage supply M14
to board D1 Main System
a4) The Board M14 might be defect.
a5) The Main System D1 might be defect.
re: re: siemens arcadis varic error
April 09, 2021 01:38
i will do
Maybe will be the D1 board
Because i habe checked all situations
Thank’s so much
April 08, 2021 10:54
Hello, my Friend!
Did you enter service browser and read error description in error log?
April 08, 2021 10:55
Hi! You must go to service and see errors. Write me on my mail if need help with service
INTUITIVE THINKING • INTUITIVE WORKS • INTUITIVE SOLUTIONS • Contact TIBS today for your GE OEC C-arm Sales, Service or Parts. Parts & systems in stock now. Exchange or sell us your old C-arm. Call 866.499.3966 or email firstname.lastname@example.org.