Rahim Rajab Ali
Siemens mammomat nova3000
October 28, 2019 10:42
Need service program manual of siemena mammomat nova 3000.
I have error code 802 to resolve.
to rate and post a comment
re: Siemens mammomat nova3000
October 29, 2019 10:37
Try swapping the Bucky's around . If the problem goes away, the problem could be the bucky switch or motor. The switch was a common problem back in the day. Siemens even came out with a kit to repair it.
October 29, 2019 10:47
Code Error 802 OKT 2 HAS RESPONDED
Presure switch on tube housing or beam form anode or door switch open
I mind your tube is active pressure switch
OKT 2 has responded
Displayed on PC - OKT pressure switch on tube housing or beam form anode or door switch.
PH1 is replaced with OKT2, HV tank with tube housing.
October 29, 2019 10:48
Er 802 “OKT 2 has responded”
Displayed on PC: OKT2 pressure switch on tube housing or beam form anode or door switch
Description Same as Er 630 (PH1 is replaced with OKT2, HV tank with tube housing).
Er 630 “Pressure switch PH1 has responded”
Displayed on PC: PH1 pressure switch on the HV tank activated
Description PH1 pressure switch on the HV tank is activated, the above-mentioned error is displayed.
Fault elimination Overload of the tube assembly or of Hl.
Line interruption (see wiring diagram).
Board D702 defective (monitoring).
November 01, 2019 11:27
Error 802 "OKT 2 has not responded"
"Okt2 pressure switch on tube housing or beam form anode or door switch"
Same as Er 630 (Replace PH1 with OKT2, hv tank with tube housing)
Overload of the tube assembly or of HI
Line interruption (see wiring diagram)
Board D702 defective (monitoring)
re: re: Siemens mammomat nova3000
November 08, 2019 10:38
For error 802, all check has done, now I am going to check board d702. All other things were fine but now I am facing error 302 and 620.
Now what should I do?
re: re: re: Siemens mammomat nova3000
November 10, 2019 11:36
“UCTRL is missing”
Displayed on PC: U
Master D702 monitors the control voltage of the transistor module in the inverter (+ 15V).
If this voltage fails, the above-mentioned error is displayed.
Fuse F5 defective (see wiring diagram).
Measure voltage on T2.
Board D702 or D710 defective.
+15V to control inverter not present
“Filament current monitoring I
Displayed on PC: I
During “Stand-by”, the master checks whether there is any filament current. If the filament
current does not reach a predetermined value, the above-mentioned error is displayed.
- filament current too low at standby
Check fuse F3 on T2 (see wiring diagram).
Check X8 connector (both on board D700 and on the cable).
Check filament wires.
Filament of tube assembly defective.
PC board D705 defective.
your tube is with trouble
re: re: re: re: Siemens mammomat nova3000
December 09, 2019 10:41
Thanks, problem has resolved but when do exposure error 801 occurred
All is well but exposure is not working.
Leading independent provider of LINAC services and repair. 99% uptime with reduced cost.