Siemens Siremobile Compact C Arm error code 7409
January 26, 2010 04:20
Siemens Siremobile Compact C Arm System. Have error code 7409.
to rate and post a comment
Re: Siemens Siremobile Compact C Arm error code 7409
January 27, 2010 08:01
Error E7409 TV-Iris wrong position
If the error 7400 to 7408 were detected, this error is displayed after each radiation release.
The user must delete this error before radiation can be released.
For more informations, read out the previous error messages.
Check the doserate adjustment.
Check the actual value potentiometer.
Check the actual value potentiometer wires.
Adepto Medical is an infusion pump rental, sales, and service company. Our focus is to deliver to our customers the best product and customer service support we can offer. Call (913) 261-9933