February 22, 2021 06:33
I have found that sp02 cable damage in the middle, cut off from both sides and tried soldering and as expected does not work. The patient monitor is Datex Ohmeda Compact S/5. Why? What I am doing wrong? I added photo below. It connects but on display shows "check probe".
Please look at pictures.
to rate and post a comment
re: SpO2 probe
February 24, 2021 11:19
Poorly soldered. The earth shell must be soldered..
Therapeutic Technologies closeout auctions take place on DOTmed.com. Bid today on 178 lots of medical equipment. Onsite or virtual inspection available. Contact Scott at +1 (856) 655-8055 or email email@example.com for an appointment.