Ziehm Exoscop 7000 Error 26 during exposure - service manual
January 11, 2011 12:40
Good morning. I need a service manual for a ziehm exoscop 7000 or some insight as to what the error code E26 represents when trying to make an exposure. I can make a 50kv manual exposure?
Is it tube arc??
Any assistance is greatly appreciated.
to rate and post a comment
re: Ziehm Exoscop 7000 Error 26 during exposure - service manual
July 12, 2012 01:46
Did you find out what the error 26 was? Im having the same problem?
re: re: Ziehm Exoscop 7000 Error 26 during exposure - service manual
November 26, 2012 12:34
No, never found out, we left it unplugged for sometime, plugged it back in and now reading error 32, which I was told means batteries on cpu board need replacing, we decided to sale as parts unit
February 27, 2013 04:43
re: re: re: Ziehm exposcop 7000 compact error code 26
July 25, 2012 08:00
It means there was a generator error, could be the generator needs to be recalibrated, or a bad connection to the tube or bad wire withing the cable harness or worse case a faulty, arcing tube. Errors 26 & 29 in most cases means the tube is arcing and needs to be replaced but you could get lucky with a recalibration or checking the connections to the tube. The high voltage is developed at the tube head which is usually at fault...good luck
Infusion Pump Repair (IPR) is a certified medical support and solutions company. We specialize in the repair, refurbishment and resale of infusion pumps. Call 1-855-477-8866