Ziehm Solo C-Arm
September 07, 2021 02:28
Ziehm in solo scope device
I get error E1719 Gain adjustment not possible. I would appreciate it if you could help me understand what it means.
to rate and post a comment
re: Ziehm Solo C-Arm
September 08, 2021 10:59
Hi Hasan, the video-signal from the CCD-camera is missing. Check cabling and CCD suply.
re: re: Ziehm Solo C-Arm
September 14, 2021 11:45
Thank you very much Jegalindof problem solved. The video cable is plugged in the wrong place. But now it gives another error.
E1226: CAN modules missing: 17.9
Do you know what this means?
At Atlas Electronics we don’t sell new monitors so we are highly motivated to repair your existing equipment. We repair all makes and models of Surgical Monitor Displays since 1981!
Call Nick at 214-923-4229 or email direct email@example.com